SMTL Medical Disposable Datacard
Blood set with pumpchamber
|All Wales Contract:||Intravenous sets and related items|
|Usage Status:||Single use|
|Commonly used name||Blood set|
Also known as a blood administration set. Transparent tubing with retractable male luer lock at distal end, a non-vented plastic spike protected by a sheath at proximal end and 2 flexible chambers in the centre to allow manual compression. The upper chamber contains a filter (180-200Âµm) and the lower contains a ball which acts as a valve. A roller clamp controls the flow rate. Sets may also include a latex-free injection site.
Key areas where used
- All clinical areas
Method of use
The roller clamp is moved into its closed position and the blood or blood product container is prepared for use. If the container is rigid, an air inlet set (see air inlet datacard) should be utilised. Using an aseptic technique, the sheath is removed from the spike and inserted into the appropriate port of the container. The container is then suspended and depending on the manufacturers instructions, the double chambers are filled accordingly and the line is primed. The luer lock connector is secured to an IV cannula. Flow is regulated using the roller clamp.
Manufacturers recommend a maximum usage time of 24 hours or use in accordance with hospital policy
Individually wrapped in a peel pouch
- Not applicable.
The following standards are relevant to this product;
|BSI||BS EN ISO 1135: part 4: 2012||Transfusion equipment for medical use|
The following table lists examples of products known to be acceptable for this product description.
|Brand Name||Product Code||Manufacturer|
|Blood administration set||RMC2071B (4mm tubing, 200um filter)||Baxter Healthcare Ltd|
P Phillips and J Ford
Please inform SMTL of any inaccuracies noted or amendments required in this datacard.
This datacard has been prepared as an aid to contracting for purchasers in the NHS in Wales, and should not be used as a substitute for literature, instructions or training from the manufacturers of the devices.
Products mentioned in this datacard are for illustration purposes only and should not be seen as an endorsement of these products.